How to use the 2 button interface video: To get the most out of the video open the link below and print out the document or open it in another window. Note the calibration coefficient Q Factor, K [P/m3] will show up on the meter as 218364 for a calibration coefficient of 218.364 on the calibration certificate.
MENU 22 Q Unit >> l/m Q Mode >> Qvw actual (We are measuring volume flow....) Q Factor >> 2121.47 Do Not Change (This is stamped on the side of the meter) Density >> 875 kg/m3 (Water is 1000) Q20mA >> 250 (FULL SCALE... If your meter is 250 l/m) Q Cut-Zero % >> 5 Q Up Al >> 250 to turn the alarm on at 100% Q Dn Al >> 25 to turn the alarm at 10% flow Damp S >> 6 Comm Address >> 0 Clear Q >> Clears Totalizer, Password = 70
MENU 33 Language >> English Pulse Type >> F_adj Pulse Factor >> 0.01 Comm Param >> 9600, No Comm Switch >> ON P_display >> MEASURE Set Pc >> 0 P0-Ref >> 101.3KPa T_display >> MEASURE Set Tc >> 20 Deg C T-TYPE >> Pt=100 Tn >> 20 Deg C Environ-T >> -20 Deg C (-10 C is fine if you do not get them cold.) Q-adj >> DO NOT change this setting
MENU 44 Signal-TYPE >> LF: DN40-500 Lqd Gain >> G=9 VPP limit >> 100mV Fre-Range >> 11=34-850Hz Work Type >> Digital an-noise TYPE >> ANS Signal-Adj-Set >> FIlter 10 or OFF 50Hz Cut >> ON K-LINE ADJ >> DO NOT change this setting
Where to get Power Supplies, Gaskets, Nuts, and Bolts Link